We are a user of the Impulse Noise Simulator INS-410/INS-420/INS-420A/INS-4310/INS-4320/INS-4320A/INS-4001/INS-4020/INS-4040. Can we use it to test the EUT operating on DC 110V?NoiseKen
No. For DC, the maximum voltage which the standard INS series simulators can handle is 60V or 65V (depending on model). For this requirement, a special (customer) order product is available.